Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. This is analogous to top down programming design in a higher level language. I have written the code but its showing Error Take note that when the instrument is turned on, the GPIB Address that is assigned to the device is displayed. Have your TA initial your sheet. So, I’m puzzled about it. What if there was a way to set up a computer program to change and measure all of the different measurements?
|Date Added:||1 February 2009|
|File Size:||9.63 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
What happens is that experiments become more efficient and take less time. Most of these defaults will work for the purposes of these experiments. Le probleme c’est que la fruence labvieq de ma boucle while a beaucoup chut et mon programme ne fonctionne plus correctement.
Would you please send the code to this forum – then we can discuss about it. To make a sinusoidal waveform use the command Sin 5.
Have your TA initial your sheet. Use the oscilloscope and the A to observe the changes of the function generator from Sine to Square to Triangular waveforms.
Firstly, that’s my mistake to not change scope setting “Probe scale” to 0. Converting Labview Test Executive 5.
Then in the next step i need to load a file into Pattern Gen module which is inbulit into one of the slot of the LAPG setup. Where the issue is? If you use some clever programming you can get the program to respond labvieew by changing a number here, or the push of a button there.
NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Re-Introduction to LabVIEW
A Sequence Box performs all instructions contained in a frame before moving on to the next frame. Do the same for the burst phase and burst period terminals on the left side and bottom of the Configure Burst node.
Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tab from 33250 top right. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation.
The way to determine exactly which device that needs to be addressed is taken care of in the programming. If the frame displayed is not Frame 0, change it to Frame 0.
The instrument is now configured. The Tools Palette is shown in Figure 5. You can then place items one by one on the panel. What if there was a way to set up a computer program to change and measure all of the different measurements? If I changed Freq, nothing changed of waveform.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
To make this section work, the device needs to have a write and then a read in that order. These are the Controls Palette and the Function Palette. Lagview Programming Interface API To help accelerate software development, all instrument drivers follow a consistent programming flow of: You can configure the Configure Burst node with several inputs.
How do I create such a file? The Keithley Multimeter takes bit more work, and the commands are 333250a.
Repeat Step 1a and 1b for the A. It can be found from Agilent website, searching for instrument drivers for AGa. I try labveiw make this parameter point to the firmware path, but the program can’t be run.
Jackie – Week 3. How to do it? The next step is optional and is not required, but could help in finding errors that occur in the labvkew with the device. This way the experimenters can concentrate on the data they are collecting instead of worrying about how everything is set up and changing dials and knobs on all the equipment.